How to remain compliant with the new regulations for workplace gas detection.
Every year there are approximately 25 fatalities from accidental carbon monoxide poisoning in England and Wales, according to data from the NHS. As carbon monoxide (CO), has no smell or taste, it is undetectable until it is too late, hence CO being dubbed, the silent killer.
After CO is inhaled, it enters the bloodstream and limits the red blood cells’ ability to carry oxygen. This lack of oxygen causes cells and tissue to die, which can prove fatal in high doses.
Due to the severity of the risk, last year EH40, Workplace exposure limits, the regulation containing the list of workplace exposure limits for use with the Control of Substances Hazardous to Health regulations was updated. The updated version reduced the alarm limit for gas, including CO readers, from 50ppm (parts per million) to 20ppm in order to increase safety.
The regulation change is EU-wide but has so far failed to gain momentum in the industry and has only been accepted and implemented by six countries including the UK and Ireland. There is an industry-wide agreement that the information relating to the standard update wasn’t distributed effectively and more could have been done to raise awareness.
We believe that the lack of information is putting workers at unnecessary risk and further information about the reduced alarm limit should be more readily available.
Change in regulation EH40 replaces the 2011 edition to take account of the new limits for substances listed in Directive 2017/164/EU.
This update means:
- Important reductions to the level of CO permitted to be present in the air from 50ppm (parts per million) to 20ppm for above ground, or 30ppm (for mining)
- This will be enforced from 2020, and in 2023 for mining due to technological challenges and associated costs
- This change will affect all gases, particularly, sulphur dioxide (SO2) and nitrogen dioxide (NO2) as they are the most frequently detected. The levels for all other gasses will vary depending on the type.
Upon the next calibration of a gas monitor, which is typically carried out every six months, the new alarm limit should automatically be updated. The onus is on the end user to send their gas monitor for recalibration and ensure the alarm limit is changed.
The frequency that a monitor needs to be recalibrated may vary by application and equipment type. Sensor life can vary depending on usage and the gas it is sensing. End users don’t always calibrate in line with manufacturers’ recommendations which means that they may not be offering the highest levels of protection in line with EH40 legislation.
As some gas detection units only have a two-year battery and sensor life, end users don’t always calibrate it as often as they should, meaning they might not be offering the highest level of protection. It’s also important for users to remember that each monitor must be bump tested before every use to ensure it’s working properly.
All new units on the market will have the limits automatically reduced, and any detector purchased after 21 August 2018 will be compliant. If unsure, it is important to seek advice. A proportion of gas monitor users are generally unaware of how to properly maintain detection products. The risks associated with gas detection are too high to be unclear.
More information on HSE's website here
Alex Turgoose is product and procurement manager at Arco
By Belinda Liversedge on 28 July 2021
The government has expanded daily contact testing with 1,200 new sites across frontline sectors, helping to avoid disruption to crucial services.
By Belinda Liversedge on 26 July 2021
93 per cent of firms plan to adopt hybrid working models, according to a Confederation of British Industry (CBI) report.
By Belinda Liversedge on 13 July 2021
Experience has taught us that we can’t guarantee people will behave responsibly to prevent Covid transmission and wear masks, the chair of the British Safety Council has warned.